Job Timeline (Ref 4626-5410)
A Timeline of this job can be seen below.
Remote Repaired awaiting Payment
Meditek Remote Repair Details
Manufacturer : Meditek
Fault Description : The buttons do not respond. We have tried changing the battery as well. The battery is also slightly loose fitting.
This chair lift controller arrived safely and on inspection the battery fitting was loose which was making a connection intermittent. The terminals were sprung back into place and then reinforced using solder on the inside of the terminal fold. This now makes a solid firm connection to the battery.
The Controller was tested thoroughly and no other issue could be replicated.
Remote Controller Testing Video
Remote Control Repair Testing Video